The PAT-T Series is a constant voltage/constant current, auto-shifting, switching DC power supply. It features a soft switching system that offers greater efficiency and lower noise. At the same time, it makes full use of high-density packaging technology to reduce the unit’s size and weight.

The chassis is the standard rack width (430 mm), and is about 130 mm (3U) high and 550 mm deep. The output power is 8 kW. Compared to series regulated products of the same capacity, it is about 1/6 the volume and 1/7 the weight. While offering a high output of 8 kW despite its small cabinet size, it also features a “power factor correction circuit.” With a high power factor of 0.95, it improves the power environment (suppresses harmonic currents) and also greatly contributes to “energy saving,” as exemplified by its simplified and miniaturized power reception and distribution modules, and overall lower power consumption.

Furthermore, an optimized heat radiation design will make operation guaranteed at ambient temperatures of up to 50 °C. It can thus be deployed in demanding applications where it must provide full-load, continuous operation despite high ambient temperatures. The layout of the operation/display panel is simple and intuitive and has been designed with viewability and usability in mind.

An RS-232C interface is provided as standard together with external analog control, monitor output, and status output connectors, enabling control from an external computer or sequencer. In addition, GPIB or USB interface can be equipped as a factory option. The unit can either be used in a standalone configuration or can be incorporated into a test system.

# Kikusui PAT-T Series

High-Efficiency, High-Capacity Switching Power Supply (Switching system, CV/CC) : 97 Models

Kategoriler: DC Güç Kaynakları, Güç Kaynakları.
Etiketler: Güç kaynakları, DC Güç kaynakları.

**Teknik Şartnameler ve İlgili Dökümanlar**

**General**

- 4 kW type can operate even with single-phase 200 volt input. (However, current is limited to about 75% of rated value.)
- Standardly equipped with RS-232C interface.
- Supports USB/GPIB/LAN interface.(Factory option)
- Controllable from Excel VBA and LabView with measuring instrument driver. Driver can be downloaded free at our web site.
- Capacity can be expanded through parallel operation (up to five units of the same model).
- Equipped with reliable output ON/OFF delay function during sequence operations.
- Memory function (three sets of voltage/current)
- Voltage/current monitor output
- Status signal output
- Remote sensing function
- Protective functions (shutdown, as well as protection against overvoltage, overcurrent, overheating, input phase interruption, fan malfunction, sensing, and bleeder circuit overheating)
- High noise resistance (for reassurance during car electronics testing)
- Good maintainability, including easy fan replacement
- 3-phase 400V input available (8kw type)

**Smart rack system (PAT-TM/TMX)**

- The large-current model assembles multiple PAT-T series units with special rack parts.
- The PAT-TMX is a model equipped with the breaker.
- A total of 82 models are available, ranging from 16 kW to 40 kW.

**Control via browser (Factory option)**

The LAN interface applies to the LXI (LAN eXtensions for Instrumentation). If that interface is used, it is possible to control and monitor the power supply from a browser.

**Car electronics applications**

- Lifetime testing of headlights, etc
- Performance and endurance testing of inverters for use in high-capacity air conditioners and motors
- Performance and endurance testing of brushless motors for use in EPS, MG, and SG units
- Performance testing of IPM, IGBT, and other power modules
- Performance testing of starter motors

**LINE UP**

Model |
Specification |

PAT20-200T |
4kW, 0 to 20V/0 to 200A |

PAT20-400T |
8kW, 0 to 20V/0 to 400A |

PAT20-800TM |
16kw,0 to 20V/0 to 800A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT20-800TMX |
16kw,0 to 20V/0 to 800A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT20-1200TM |
24kw,0 to 20V/0 to 1200A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT20-1200TMX |
24kw,0 to 20V/0 to 1200A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT20-1600TM |
32kw,0 to 20V/0 to 1600A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT20-1600TMX |
32kw,0 to 20V/0 to 1600A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT20-2000TM |
40kw,0 to 20V/0 to 2000A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT20-2000TMX |
40kw,0 to 20V/0 to 2000A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT30-266T |
8kW, 0 to 30V/0 to 266A |

PAT30-532TM |
16kw,0 to 30V/0 to 532A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT30-532TMX |
16kw,0 to 30V/0 to 532A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT30-798TM |
24kw,0 to 30V/0 to 798A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT30-798TMX |
24kw,0 to 30V/0 to 798A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT30-1064TM |
32kw,0 to 30V/0 to 1064A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT30-1064TMX |
32kw,0 to 30V/0 to 1064A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT30-1330TM |
40kw,0 to 30V/0 to 1330A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT30-1330TMX |
40kw,0 to 30V/0 to 1330A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT40-100T |
4kW, 0 to 40V/0 to 100A |

PAT40-200T |
8kW, 0 to 40V/0 to 200A |

PAT40-400TM |
16kw,0 to 40V/0 to 400A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT40-400TMX |
16kw,0 to 40V/0 to 400A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT40-600TM |
24kw,0 to 40V/0 to 600A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT40-600TMX |
24kw,0 to 40V/0 to 600A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT40-800TM |
32kw,0 to 40V/0 to 800A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT40-800TMX |
32kw,0 to 40V/0 to 800A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT40-1000TM |
40kw,0 to 40V/0 to 1000A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT40-1000TMX |
40kw,0 to 40V/0 to 1000A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT60-67T |
4kW, 0 to 60V/0 to 67A |

PAT60-133T |
8kW, 0 to 60V/0 to 133A |

PAT60-266TM |
16kw,0 to 60V/0 to 266A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT60-266TMX |
16kw,0 to 60V/0 to 266A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT60-399TM |
24kw,0 to 60V/0 to 399A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT60-399TMX |
24kw,0 to 60V/0 to 399A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT60-532TM |
32kw,0 to 60V/0 to 532A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT60-532TMX |
32kw,0 to 60V/0 to 532A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT60-665TM |
40kw,0 to 60V/0 to 665A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT60-665TMX |
40kw,0 to 60V/0 to 665A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT80-100T |
8kW, 0 to 80V/0 to 100A |

PAT80-200TM |
16kw,0 to 80V/0 to 200A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT80-200TMX |
16kw,0 to 80V/0 to 200A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT80-300TM |
24kw,0 to 80V/0 to 300A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT80-300TMX |
24kw,0 to 80V/0 to 300A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT80-400TM |
32kw,0 to 80V/0 to 400A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT80-400TMX |
32kw,0 to 80V/0 to 400A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT80-500TM |
40kw,0 to 80V/0 to 500A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT80-500TMX |
40kw,0 to 80V/0 to 500A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT160-25T |
4kW, 0 to 160V/0 to 25A |

PAT160-50T |
8kW, 0 to 160V/0 to 50A |

PAT160-100TM |
16kw,0 to 160V/0 to 100A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT160-100TMX |
16kw,0 to 160V/0 to 100A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT160-150TM |
24kw,0 to 160V/0 to 150A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT160-150TMX |
24kw,0 to 160V/0 to 150A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT160-200TM |
32kw,0 to 160V/0 to 200A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT160-200TMX |
32kw,0 to 160V/0 to 200A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT160-250TM |
40kw,0 to 160V/0 to 250A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT160-250TMX |
40kw,0 to 160V/0 to 250A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT250-32T |
8kW, 0 to 250V/0 to 32A |

PAT250-64TM |
16kw,0 to 250V/0 to 64A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT250-64TMX |
16kw,0 to 250V/0 to 64A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT250-96TM |
24kw,0 to 250V/0 to 96A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT250-96TMX |
24kw,0 to 250V/0 to 96A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT250-128TM |
32kw,0 to 250V/0 to 128A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT250-128TMX |
32kw,0 to 250V/0 to 128A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT250-160TM |
40kw,0 to 250V/0 to 160A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT250-160TMX |
40kw,0 to 250V/0 to 160A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT350-22.8T |
8kW, 0 to 350V/0 to 22.8A |

PAT350-45.6TM |
16kw,0 to 350V/0 to 45.6A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT350-45.6TMX |
16kw,0 to 350V/0 to 45.6A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT350-68.4TM |
24kw,0 to 350V/0 to 68.4A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT350-68.4TMX |
24kw,0 to 350V/0 to 68.4A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT350-91.2TM |
32kw,0 to 350V/0 to 91.2A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT350-91.2TMX |
32kw,0 to 350V/0 to 91.2A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT350-114TM |
40kw,0 to 350V/0 to 114A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT350-114TMX |
40kw,0 to 350V/0 to 114A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT500-16T |
8kW, 0 to 500V/0 to 16A |

PAT500-32TM |
16kw,0 to 500V/0 to 32A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT500-32TMX |
16kw,0 to 500V/0 to 32A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT500-48TM |
24kw,0 to 500V/0 to 48A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT500-48TMX |
24kw,0 to 500V/0 to 48A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT500-64TM |
32kw,0 to 500V/0 to 64A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT500-64TMX |
32kw,0 to 500V/0 to 64A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT500-80TM |
40kw,0 to 500V/0 to 80A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT500-80TMX |
40kw,0 to 500V/0 to 80A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT650-12.3T |
8kW, 0 to 650V/0 to 12.3A |

PAT650-24.6TM |
16kw,0 to 650V/0 to 24.6A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT650-24.6TMX |
16kw,0 to 650V/0 to 24.6A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT650-36.9TM |
24kw,0 to 650V/0 to 36.9A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT650-36.9TMX |
24kw,0 to 650V/0 to 36.9A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT650-49.2TM |
32kw,0 to 650V/0 to 49.2A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT650-49.2TMX |
32kw,0 to 650V/0 to 49.2A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT650-61.5TM |
40kw,0 to 650V/0 to 61.5A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT650-61.5TMX |
40kw,0 to 650V/0 to 61.5A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

PAT850-9.4T |
8kW, 0 to 850V/0 to 9.4A |

PAT850-18.8TM |
16kw,0 to 850V/0 to 18.8A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.) |

PAT850-18.8TMX |
16kw,0 to 850V/0 to 18.8A, Smart rack system (The large-current model assembles multiple PAT-T series units with special rack parts.), Model with breaker |

**8 KW Type Specifications**

Item | PAT20-400T | PAT30-266T | PAT40-200T | PAT60-133T | PAT80-100T | |||

Input | Nominal input rated voltage | Three-phase 200 to 240 VAC, 50-60 Hz | ||||||

Input voltage range/Input frequency range | 180 V to 250 V / 47Hz to 63 Hz | |||||||

Efficiency | 85% (min) [at input voltage of 200 VAC and rated load] | |||||||

Power factor | 0.95 (typical) [at input voltage of 200 VAC and rated load] | |||||||

Input current | 32 A (max) [rated load] | |||||||

Inrush current | 100 A peak (max) | |||||||

Input power | 10kVA (max) | |||||||

Output | Rating | Rated output power | 8 kW | |||||

Rated output voltage | 20.00 V | 30.00 V | 40.00 V | 60.0 V | 80.0 V | |||

Rated output current | 400.0 A | 266.0 A | 200.0 A | 133.0 A | 100.0 A | |||

Constant voltage | Setting accuracy | ± (0.2% of rating +50 mV) | ||||||

Max setting voltage | 105% of rating | |||||||

Line requlation | ± (0.05% of rating +5 mV) | |||||||

Load requlation | ± (0.1% of rating +5 mV) | |||||||

Transient response time | 5 ms (at an instantaneous change in the load current from 50% to 100%) | |||||||

Ripple noise | 100 mVp-p | 300 mVp-p | 300 mVp-p | 350 mVp-p | 350 mVp-p | |||

When the measurement frequency band is 10 Hz to 20 MHz | ||||||||

10 mVrms | 20 mVrms | 30 mVrms | 30 mVrms | 30 mVrms | ||||

When the measurement frequency band is 5 Hz to 1 MHz | ||||||||

Raise time | 100 ms (rated load)/100 ms (no load) | |||||||

Fall time | 100 ms (rated load)/2000 ms (no load) | |||||||

Temperature coefficient | 100 ppm/˚C (max) [with external analog control] | |||||||

Constant current* | Setting accuracy | ± (0.5% of rating +50 mA) | ||||||

Max setting current | 105% of rating | |||||||

Line requlation | ± (0.1% of rating +30 mA) | |||||||

Load requlation | ± (0.2% of rating +30 mA) | |||||||

Ripple noise | 500 mArms | 400 mArms | 400 mArms | 350 mArms | 300 mArms | |||

When the measurement frequency band is 5 Hz to 1 MHz | ||||||||

Temperature coefficient | 200 ppm/˚C (typ) [with external analog control] | |||||||

OUTPUT ON/OFF delay | OFF. 0.1 to 10.0 s (resolution: 0.1 s) | |||||||

Voltage display | Maximum display | 99.99 | ||||||

Error | ± (0.2% of reading +5 digits) at 23˚C ±5˚C | |||||||

Current display | Maximum display | 999.9 | ||||||

Error | ± (0.5% of reading +5 digits) at 23˚C ±5˚C | |||||||

Protection function | Overvoltage protection (OVP) / Overcurrent protection (OCP) / Overheat protection (OHP) /Input open phase protection (PHASE) / Fan error protection (FAN) / Mis-connection protection (SENSE) / Breeder circuit overheat protection (BOHP) / Shutdown (SD) | |||||||

External analog control | OUTPUT ON/OFF control, etc. | OUTPUT ON/OFF, SHUTDOWN | ||||||

Constant voltage, external voltage control | 0% to 100% of the rated output voltage at 0 to 10 V | |||||||

Constant voltage, external resistance control | 0% to 100% or 100% to 0% of the rated output voltage at 0 Wto 10 kW | |||||||

Constant current, external voltage control | 0% to 100% of tared output current at 0 to 10 V | |||||||

Constant current, external resistance control | 0% to 100% or 100% to 0% of rated output current at 0 Wto 10 kW | |||||||

Monitor output | Output voltage | 10.00 V ±0.25 V at rated voltage output | ||||||

0.00 V ±0.25 V at 0 V output | ||||||||

Output current | 10.00 V ±0.25 V at rated current output | |||||||

0.00 V ±0.25 V at 0 A current | ||||||||

Status output | OUT ON, CV, CC, ALARM, POWER ON, POWER OFF, insulated open collector | |||||||

Remote control | Equipped with RS-232C interface as standard. SCPI commands, up to 38,400 bps | |||||||

Operating temperature/humidity range | 0˚C to 50˚C, 20% to 85% rh | |||||||

Storage temperature/humidity range | -25˚C to 70˚C, 90% rh or less (non-condensing) | |||||||

Dimensions (maximum) | 430 (440)(16.93''(17.32'')) W × 129.2 (155)(5.09''(6.10'')) H × 550 (620)(21.65''(24.41'')) D mm | |||||||

Weight | Approx. 26 kg (57.32 Ib) | Approx. 27 kg (59.52 Ib) | Approx. 25 kg (55.12 Ib) | Approx. 24 kg (52.91 Ib) |

Item | PAT160-50T | PAT250-32T | PAT350-22.8T | PAT500-16T | PAT650-12.3T | PAT850-9.4T | |||

Input |
Nominal input rated voltage | Three-phase 200 to 240 VAC, 50-60 Hz | |||||||

Input voltage range/Input frequency range | 180 V to 250 V / 47Hz to 63 Hz | ||||||||

Efficiency | 85% (min) [at input voltage of 200 VAC and rated load] | ||||||||

Power factor | 0.95 (typical) [at input voltage of 200 VAC and rated load] | ||||||||

Input current | 32 A (max) [rated load] | ||||||||

Inrush current | 100 A peak (max) | ||||||||

Input power | 10kVA (max) | ||||||||

Output |
Rating |
Rated output power | 8 kW | ||||||

Rated output voltage | 160.0 V | 250.0 V | 350.0 V | 500.0 V | 650.0 V | 850.0 V | |||

Rated output current | 50.0 A | 32.0 A | 22.8 A | 16.0 A | 12.3 A | 9.4 A | |||

Constant voltage |
Setting accuracy | ± (0.2% of rating +50 mV) | |||||||

Max setting voltage | 105% of rating | ||||||||

Line requlation | ± (0.05% of rating +5 mV) | ||||||||

Load requlation | ± (0.1% of rating +5 mV) | ||||||||

Transient response time | 5 ms (with sensing at external output, at an instantaneous change in the load current from 50% to 100%) | ||||||||

Ripple noise |
350 mVp-p | 450 mVp-p | 450 mVp-p | 600 mVp-p | 600 mVp-p | 600 mVp-p | |||

When the measurement frequency band is 10 Hz to 20 MHz | |||||||||

30 mVrms | 50 mVrms | 50 mVrms | 100 mVrms | 100 mVrms | 100 mVrms | ||||

When the measurement frequency band is 5 Hz to 1 MHz | |||||||||

Raise time | 100 ms (rated load)/100 ms (no load) | ||||||||

Fall time | 100 ms (rated load)/2000 ms (no load) | 200 ms (rated load)/ 4000 ms (no load) | |||||||

Temperature coefficient | 100 ppm/˚C (max) [with external analog control] | ||||||||

Constant current* |
Setting accuracy | ± (0.5% of rating +50 mA) | ± (1% of rating +100 mA) | ||||||

Max setting current | 105% of rating | ||||||||

Line requlation | ± (0.1% of rating +30 mA) | ||||||||

Load requlation | ± (0.2% of rating +30 mA) | ||||||||

Ripple noise |
200 mArms | 200 mArms | 200 mArms | 200 mArms | 150 mArms | 120 mArms | |||

When the measurement frequency band is 5 Hz to 1 MHz | |||||||||

Temperature coefficient | 200 ppm/˚C (typ) [with external analog control] | ||||||||

OUTPUT ON/OFF delay | OFF. 0.1 to 10.0 s (resolution: 0.1 s) | ||||||||

Voltage display |
Maximum display | 999.9 | |||||||

Error | ± (0.2% of reading +5 digits) at 23˚C ±5˚C | ||||||||

Current display |
Maximum display | 999.9 | 99.99 | ||||||

Error | ± (0.5% of reading +5 digits) at 23˚C ±5˚C | ||||||||

Protection function |
Overvoltage protection (OVP) / Overcurrent protection (OCP) / Overheat protection (OHP) /Input open phase protection (PHASE) / Fan error protection (FAN) / Mis-connection protection (SENSE) / Breeder circuit overheat protection (BOHP) / Shutdown (SD) | ||||||||

External analog control |
OUTPUT ON/OFF control, etc. | OUTPUT ON/OFF, SHUTDOWN | |||||||

Constant voltage, external voltage control | 0% to 100% of the rated output voltage at 0 to 10 V | ||||||||

Constant voltage, external resistance control | 0% to 100% or 100% to 0% of the rated output voltage at 0 Wto 10 kW | ||||||||

Constant current, external voltage control | 0% to 100% of tared output current at 0 to 10 V | ||||||||

Constant current, external resistance control | 0% to 100% or 100% to 0% of rated output currenn at 0 Wto 10 kW | ||||||||

Monitor output |
Output voltage |
10.00 V ±0.25 V at rated voltage output | |||||||

0.00 V ±0.25 V at 0 V output | |||||||||

Output current |
10.00 V ±0.25 V at rated current output | ||||||||

0.00 V ±0.25 V at 0 A current | |||||||||

Status output | OUT ON, CV, CC, ALARM, POWER ON, POWER OFF, insulated open collector | ||||||||

Remote control | Equipped with RS-232C interface as standard. SCPI commands, up to 38,400 bps | ||||||||

Operating temperature/humidity range | 0˚C to 50˚C, 20% to 85% rh | ||||||||

Storage temperature/humidity range | -25˚C to 70˚C, 90% rh or less (non-condensing) | ||||||||

Dimensions (maximum) | 430 (440)(16.93''(17.32'')) W × 129.2 (155)(5.09''(6.10'')) H × 550 (620)(21.65''(24.41'')) D mm | ||||||||

Weight | Approx. 24 kg (52.91 Ib) | Approx. 23 kg (50.71 Ib) | Approx. 22 kg (48.50 Ib) | Approx. 23 kg (50.71 Ib) |

**4 kW Type Specifications**

Item | PAT20-200T | PAT40-100T | PAT60-67T | PAT160-25T | |||

Input |
Nominal input rated voltage | Single-phase/three-phase 200 to 240 VAC, 50-60 Hz | |||||

Input voltage range/Input frequency range | 180 V to 250 V / 47 Hz to 63 Hz | ||||||

Efficiency | 84% (min) | 85% (min) [at input voltage of 200 VAC and rated load] | |||||

Power factor | 0.95 (typical) [at input voltage of 200 VAC andratedload] s |
||||||

Input current | Single-phase 22 A (max) [at 3 kW load]/three-phase 17 A (max) [at rated load] | ||||||

Inrush current | 50 A peak (max) | ||||||

Input power | Single-phase 4 kVA (max) [at 3 kW load]/three-phase 5 kVA (max) [at rated load] | ||||||

Output |
Rating |
Rated output power | 4 kW(three-phase input mode) / 3 kW(single-phase input mode) | ||||

Rated output voltage | 20.00 V | 40.00 V | 60.00 V | 160.0 V | |||

Rated output current | 200.0 A | 100.0 A | 67.00 A | 25.00 A | |||

Constant voltage |
Setting accuracy | ± (0.2% of rating +50 mV) | |||||

Max setting voltage | 105% of rating | ||||||

Line requlation | ± (0.05% of rating +5 mV) | ||||||

Load requlation | ± (0.1% of rating +5 mV) | ||||||

Transient response time | 5 ms (at instantaneous change in load current from 50% to 100%) | ||||||

Ripple noise |
100 mVp-p | 300m Vp-p | 350 mVp-p | 350 mVp-p | |||

When the measurement frequency band is 10 Hz to 20 MHz | |||||||

10 mVrms | 30 mVrms | 30 mVrms | 30 mVrms | ||||

When the measurement frequency band is 5 Hz to 1 MHz | |||||||

Raise time | 100 ms (rated load)/100 ms (no load) | ||||||

Fall time | 100 ms (rated load)/2000 ms (no load) | ||||||

Temperature coefficient | 100 ppm/˚C (max) [with external analog control] | ||||||

Constant current* |
Setting accuracy | ± (0.5% of rating +50 mA) | |||||

Max setting current | 105% of rating × 75% (with single-phase input) / 105% of rating (with three-phase input) | ||||||

Line requlation | ± (0.1% of rating +30 mA) | ||||||

Load requlation | ± (0.2% of rating +30 mA) | ||||||

Ripple noise |
400 mArms | 300 mArms | 250 mArms | 200 mArms | |||

When the measurement frequency band is 5 Hz to 1 MHz | |||||||

Temperature coefficient | 200 ppm/˚C (typ) [with external analog control] | ||||||

OUTPUT ON/OFF delay | OFF. 0.1 to 10.0 s (resolution: 0.1 s) | ||||||

Voltage display |
Maximum display | 99.99 | 999.9 | ||||

Error | ± (0.2% of reading +5 digits) at 23˚C ±5˚C | ||||||

Current display |
Maximum display | 999.9 | 99.99 | ||||

Error | ± (0.5% of reading +5 digits) at 23˚C ±5˚C | ||||||

Protection function |
Overvoltage protection (OVP) / Overcurrent protection (OCP) / Overheat protection (OHP) / Input open phase protection (PHASE) / Fan error protection (FAN) /Mis-connection protection (SENSE) / Breeder circuit overheat protection (BOHP) / Shutdown (SD) | ||||||

External analog control |
OUTPUT ON/OFF control, etc. | OUTPUT ON/OFF, SHUTDOWN | |||||

Constant voltage, external voltage control | 0% to 100% of the rated output voltage at 0 to 10 V | ||||||

Constant voltage, external resistance control | 0% to 100% or 100% to 0% of the rated output voltage at 0 Wto 10 kW | ||||||

Constant current, external voltage control | 0% to 100% of tared output current at 0 to 10 V | ||||||

Constant current, external resistance control | 0% to 100% or 100% to 0% of rated output currenn at 0 Wto 10 kW | ||||||

Monitor output |
Output voltage |
10.00 V ±0.25 V at rated voltage output | |||||

0.00 V ±0.25 V at 0 V output | |||||||

Output current |
10.00 V ±0.25 V at rated current output | ||||||

0.00 V ±0.25 V at 0 A current | |||||||

Status output | OUT ON, CV, CC, ALARM, POWER ON, POWER OFF, insulated open collector | ||||||

Remote control | Equipped with RS-232C interface as standard. SCPI commands, up to 38,400 bps | ||||||

Operating temperature/humidity range | 0˚C to 50˚C, 20% to 85% rh | ||||||

Storage temperature/humidity range | -25˚C to 70˚C, 90% rh or less (non-condensing) | ||||||

Dimensions (maximum) | 430 (440)(16.93''(17.32'')) W × 129.2 (155)(5.09''(6.10'')) H × 550 (620)(21.65''(24.41'')) D mm | ||||||

Weight | Approx. 20 kg(44.09 Ib) | Approx. 19 kg(41.89 Ib) | Approx. 18 kg(39.68 Ib) |

**OPTIONS**

**PC01-PAT**

Parallel operation cable for PAT series length:250 mm

**AC8-4P4M-M6C**

Input Power cable, Three-phase, 4-conductor 4m M6

**KRB3-TOS**

Option(s) for Electrical Safety Testers. Bracket (Inch size) for TOS9200, TOS9201, TOS8870A, PAM(2kW type), PCR1000M, LZ1205W, PLZ-4W/4WH, PWR, PAT, PBZ

**KRB150-TOS**

Option(s) for Electrical Safety Testers. Bracket (Metric size) for TOS9200, TOS9201, TOS8870A, PAM(2kW type), PCR1000M, PLZ1205W, PLZ-4W/4WH, PWR, PAT, PBZ

**OP01-PAT**

Power switch guard

**VS01**

Vertical stand

### Ürünler

- ►Solar Simülatör
- ►Sayaç Test Cihazları
- ►Motor Test Cihazları
- Yüksek Gerilim Bolücüler
- Yüksek Akım Kaynağı
- Megohm meter
- ►Toprak Megerleri
- ►İzolasyon/Yüksek Gerilim (Delinme-Hipot)/Kaçak Akım/Topraklama Test Cihazları
- ►Güç Analizörleri
- ►Güneş Ölçümü Test Cihazları
- ►Trafo Test Cihazları
- ►Mili-Mikro Ohm Metreler
- ►Şalt Sahası Test Cihazları
- ►Kablo Test Cihazları
- ►Termal Kameralar
- ►Kalibratörler
- ►LCR Metreler
- ►Osiloskoplar
- ►Data Loggerlar
- ▼Güç Kaynakları
- ►PD (Kısmi Boşalma) Tespit Cihazları
- ►Akü Test Cihazları
- ►Diğer Ürünler
- ►Deney Setleri / Eğitim Setleri
- Yeşil Enerji Eğitim Setleri
- Akıllı Bina Deney Setleri (EIB/KNX)
- Fonksiyon/Arbitary Dalga Formu Jernaratörü
- Elektrik Tesisatı Eğitim Setleri
- Flicker Eğitim Seti
- Temel Elektrik Mühendisliği/Elektronik/Analog teknolojisi
- Güç Elektroniği / Kontrol Mühendisliği Deney Setleri
- Elektrik Makineleri Eğitim Seti
- Haberleşme / Radyo Mühendisliği Eğitim Setleri
- Dijital ve Otomasyon Teknolojisi Eğitim Setleri
- Kontaktör Kontrol Eğitim Setleri
- Mekatronik Kompakt Sistem Eğitim Seti

- ►İş Güvenliği Ekipmanları ve Test Cihazları
- ►Laboratuvar & Çevresel Ölçüm Cihazları
- ►Multimetreler
- ►Pens Ampermetreler
- ►Röle Test Cihazları
- ►Voltaj Dedektörleri
- ►Yarı İletken Test ve Üretim Sistemleri
- ►Piranometreler
- ►LPCVD/PECVD
- ►PAR / FAR Ölçümü