The PATT Series is a constant voltage/constant current, autoshifting, switching DC power supply. It features a soft switching system that offers greater efficiency and lower noise. At the same time, it makes full use of highdensity packaging technology to reduce the unit’s size and weight.
The chassis is the standard rack width (430 mm), and is about 130 mm (3U) high and 550 mm deep. The output power is 8 kW. Compared to series regulated products of the same capacity, it is about 1/6 the volume and 1/7 the weight. While offering a high output of 8 kW despite its small cabinet size, it also features a “power factor correction circuit.” With a high power factor of 0.95, it improves the power environment (suppresses harmonic currents) and also greatly contributes to “energy saving,” as exemplified by its simplified and miniaturized power reception and distribution modules, and overall lower power consumption.
Furthermore, an optimized heat radiation design will make operation guaranteed at ambient temperatures of up to 50 °C. It can thus be deployed in demanding applications where it must provide fullload, continuous operation despite high ambient temperatures. The layout of the operation/display panel is simple and intuitive and has been designed with viewability and usability in mind.
An RS232C interface is provided as standard together with external analog control, monitor output, and status output connectors, enabling control from an external computer or sequencer. In addition, GPIB or USB interface can be equipped as a factory option. The unit can either be used in a standalone configuration or can be incorporated into a test system.
Kikusui PATT Series
HighEfficiency, HighCapacity Switching Power Supply (Switching system, CV/CC) : 97 Models
General
 4 kW type can operate even with singlephase 200 volt input. (However, current is limited to about 75% of rated value.)
 Standardly equipped with RS232C interface.
 Supports USB/GPIB/LAN interface.(Factory option)
 Controllable from Excel VBA and LabView with measuring instrument driver. Driver can be downloaded free at our web site.
 Capacity can be expanded through parallel operation (up to five units of the same model).
 Equipped with reliable output ON/OFF delay function during sequence operations.
 Memory function (three sets of voltage/current)
 Voltage/current monitor output
 Status signal output
 Remote sensing function
 Protective functions (shutdown, as well as protection against overvoltage, overcurrent, overheating, input phase interruption, fan malfunction, sensing, and bleeder circuit overheating)
 High noise resistance (for reassurance during car electronics testing)
 Good maintainability, including easy fan replacement
 3phase 400V input available (8kw type)
Smart rack system (PATTM/TMX)
 The largecurrent model assembles multiple PATT series units with special rack parts.
 The PATTMX is a model equipped with the breaker.
 A total of 82 models are available, ranging from 16 kW to 40 kW.
Control via browser (Factory option)
The LAN interface applies to the LXI (LAN eXtensions for Instrumentation). If that interface is used, it is possible to control and monitor the power supply from a browser.
Car electronics applications
 Lifetime testing of headlights, etc
 Performance and endurance testing of inverters for use in highcapacity air conditioners and motors
 Performance and endurance testing of brushless motors for use in EPS, MG, and SG units
 Performance testing of IPM, IGBT, and other power modules
 Performance testing of starter motors
LINE UP
Model  Specification 
PAT20200T  4kW, 0 to 20V/0 to 200A 
PAT20400T  8kW, 0 to 20V/0 to 400A 
PAT20800TM  16kw,0 to 20V/0 to 800A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT20800TMX  16kw,0 to 20V/0 to 800A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT201200TM  24kw,0 to 20V/0 to 1200A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT201200TMX  24kw,0 to 20V/0 to 1200A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT201600TM  32kw,0 to 20V/0 to 1600A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT201600TMX  32kw,0 to 20V/0 to 1600A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT202000TM  40kw,0 to 20V/0 to 2000A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT202000TMX  40kw,0 to 20V/0 to 2000A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT30266T  8kW, 0 to 30V/0 to 266A 
PAT30532TM  16kw,0 to 30V/0 to 532A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT30532TMX  16kw,0 to 30V/0 to 532A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT30798TM  24kw,0 to 30V/0 to 798A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT30798TMX  24kw,0 to 30V/0 to 798A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT301064TM  32kw,0 to 30V/0 to 1064A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT301064TMX  32kw,0 to 30V/0 to 1064A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT301330TM  40kw,0 to 30V/0 to 1330A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT301330TMX  40kw,0 to 30V/0 to 1330A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT40100T  4kW, 0 to 40V/0 to 100A 
PAT40200T  8kW, 0 to 40V/0 to 200A 
PAT40400TM  16kw,0 to 40V/0 to 400A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT40400TMX  16kw,0 to 40V/0 to 400A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT40600TM  24kw,0 to 40V/0 to 600A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT40600TMX  24kw,0 to 40V/0 to 600A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT40800TM  32kw,0 to 40V/0 to 800A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT40800TMX  32kw,0 to 40V/0 to 800A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT401000TM  40kw,0 to 40V/0 to 1000A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT401000TMX  40kw,0 to 40V/0 to 1000A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT6067T  4kW, 0 to 60V/0 to 67A 
PAT60133T  8kW, 0 to 60V/0 to 133A 
PAT60266TM  16kw,0 to 60V/0 to 266A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT60266TMX  16kw,0 to 60V/0 to 266A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT60399TM  24kw,0 to 60V/0 to 399A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT60399TMX  24kw,0 to 60V/0 to 399A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT60532TM  32kw,0 to 60V/0 to 532A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT60532TMX  32kw,0 to 60V/0 to 532A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT60665TM  40kw,0 to 60V/0 to 665A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT60665TMX  40kw,0 to 60V/0 to 665A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT80100T  8kW, 0 to 80V/0 to 100A 
PAT80200TM  16kw,0 to 80V/0 to 200A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT80200TMX  16kw,0 to 80V/0 to 200A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT80300TM  24kw,0 to 80V/0 to 300A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT80300TMX  24kw,0 to 80V/0 to 300A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT80400TM  32kw,0 to 80V/0 to 400A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT80400TMX  32kw,0 to 80V/0 to 400A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT80500TM  40kw,0 to 80V/0 to 500A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT80500TMX  40kw,0 to 80V/0 to 500A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT16025T  4kW, 0 to 160V/0 to 25A 
PAT16050T  8kW, 0 to 160V/0 to 50A 
PAT160100TM  16kw,0 to 160V/0 to 100A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT160100TMX  16kw,0 to 160V/0 to 100A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT160150TM  24kw,0 to 160V/0 to 150A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT160150TMX  24kw,0 to 160V/0 to 150A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT160200TM  32kw,0 to 160V/0 to 200A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT160200TMX  32kw,0 to 160V/0 to 200A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT160250TM  40kw,0 to 160V/0 to 250A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT160250TMX  40kw,0 to 160V/0 to 250A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT25032T  8kW, 0 to 250V/0 to 32A 
PAT25064TM  16kw,0 to 250V/0 to 64A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT25064TMX  16kw,0 to 250V/0 to 64A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT25096TM  24kw,0 to 250V/0 to 96A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT25096TMX  24kw,0 to 250V/0 to 96A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT250128TM  32kw,0 to 250V/0 to 128A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT250128TMX  32kw,0 to 250V/0 to 128A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT250160TM  40kw,0 to 250V/0 to 160A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT250160TMX  40kw,0 to 250V/0 to 160A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT35022.8T  8kW, 0 to 350V/0 to 22.8A 
PAT35045.6TM  16kw,0 to 350V/0 to 45.6A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT35045.6TMX  16kw,0 to 350V/0 to 45.6A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT35068.4TM  24kw,0 to 350V/0 to 68.4A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT35068.4TMX  24kw,0 to 350V/0 to 68.4A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT35091.2TM  32kw,0 to 350V/0 to 91.2A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT35091.2TMX  32kw,0 to 350V/0 to 91.2A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT350114TM  40kw,0 to 350V/0 to 114A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT350114TMX  40kw,0 to 350V/0 to 114A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT50016T  8kW, 0 to 500V/0 to 16A 
PAT50032TM  16kw,0 to 500V/0 to 32A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT50032TMX  16kw,0 to 500V/0 to 32A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT50048TM  24kw,0 to 500V/0 to 48A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT50048TMX  24kw,0 to 500V/0 to 48A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT50064TM  32kw,0 to 500V/0 to 64A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT50064TMX  32kw,0 to 500V/0 to 64A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT50080TM  40kw,0 to 500V/0 to 80A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT50080TMX  40kw,0 to 500V/0 to 80A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT65012.3T  8kW, 0 to 650V/0 to 12.3A 
PAT65024.6TM  16kw,0 to 650V/0 to 24.6A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT65024.6TMX  16kw,0 to 650V/0 to 24.6A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT65036.9TM  24kw,0 to 650V/0 to 36.9A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT65036.9TMX  24kw,0 to 650V/0 to 36.9A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT65049.2TM  32kw,0 to 650V/0 to 49.2A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT65049.2TMX  32kw,0 to 650V/0 to 49.2A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT65061.5TM  40kw,0 to 650V/0 to 61.5A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT65061.5TMX  40kw,0 to 650V/0 to 61.5A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
PAT8509.4T  8kW, 0 to 850V/0 to 9.4A 
PAT85018.8TM  16kw,0 to 850V/0 to 18.8A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.) 
PAT85018.8TMX  16kw,0 to 850V/0 to 18.8A, Smart rack system (The largecurrent model assembles multiple PATT series units with special rack parts.), Model with breaker 
8 KW Type Specifications
Item  PAT20400T  PAT30266T  PAT40200T  PAT60133T  PAT80100T  
Input  Nominal input rated voltage  Threephase 200 to 240 VAC, 5060 Hz  
Input voltage range/Input frequency range  180 V to 250 V / 47Hz to 63 Hz  
Efficiency  85% (min) [at input voltage of 200 VAC and rated load]  
Power factor  0.95 (typical) [at input voltage of 200 VAC and rated load]  
Input current  32 A (max) [rated load]  
Inrush current  100 A peak (max)  
Input power  10kVA (max)  
Output  Rating  Rated output power  8 kW  
Rated output voltage  20.00 V  30.00 V  40.00 V  60.0 V  80.0 V  
Rated output current  400.0 A  266.0 A  200.0 A  133.0 A  100.0 A  
Constant voltage  Setting accuracy  ± (0.2% of rating +50 mV)  
Max setting voltage  105% of rating  
Line requlation  ± (0.05% of rating +5 mV)  
Load requlation  ± (0.1% of rating +5 mV)  
Transient response time  5 ms (at an instantaneous change in the load current from 50% to 100%)  
Ripple noise  100 mVpp  300 mVpp  300 mVpp  350 mVpp  350 mVpp  
When the measurement frequency band is 10 Hz to 20 MHz  
10 mVrms  20 mVrms  30 mVrms  30 mVrms  30 mVrms  
When the measurement frequency band is 5 Hz to 1 MHz  
Raise time  100 ms (rated load)/100 ms (no load)  
Fall time  100 ms (rated load)/2000 ms (no load)  
Temperature coefficient  100 ppm/˚C (max) [with external analog control]  
Constant current*  Setting accuracy  ± (0.5% of rating +50 mA)  
Max setting current  105% of rating  
Line requlation  ± (0.1% of rating +30 mA)  
Load requlation  ± (0.2% of rating +30 mA)  
Ripple noise  500 mArms  400 mArms  400 mArms  350 mArms  300 mArms  
When the measurement frequency band is 5 Hz to 1 MHz  
Temperature coefficient  200 ppm/˚C (typ) [with external analog control]  
OUTPUT ON/OFF delay  OFF. 0.1 to 10.0 s (resolution: 0.1 s)  
Voltage display  Maximum display  99.99  
Error  ± (0.2% of reading +5 digits) at 23˚C ±5˚C  
Current display  Maximum display  999.9  
Error  ± (0.5% of reading +5 digits) at 23˚C ±5˚C  
Protection function  Overvoltage protection (OVP) / Overcurrent protection (OCP) / Overheat protection (OHP) /Input open phase protection (PHASE) / Fan error protection (FAN) / Misconnection protection (SENSE) / Breeder circuit overheat protection (BOHP) / Shutdown (SD)  
External analog control  OUTPUT ON/OFF control, etc.  OUTPUT ON/OFF, SHUTDOWN  
Constant voltage, external voltage control  0% to 100% of the rated output voltage at 0 to 10 V  
Constant voltage, external resistance control  0% to 100% or 100% to 0% of the rated output voltage at 0 Wto 10 kW  
Constant current, external voltage control  0% to 100% of tared output current at 0 to 10 V  
Constant current, external resistance control  0% to 100% or 100% to 0% of rated output current at 0 Wto 10 kW  
Monitor output  Output voltage  10.00 V ±0.25 V at rated voltage output  
0.00 V ±0.25 V at 0 V output  
Output current  10.00 V ±0.25 V at rated current output  
0.00 V ±0.25 V at 0 A current  
Status output  OUT ON, CV, CC, ALARM, POWER ON, POWER OFF, insulated open collector  
Remote control  Equipped with RS232C interface as standard. SCPI commands, up to 38,400 bps  
Operating temperature/humidity range  0˚C to 50˚C, 20% to 85% rh  
Storage temperature/humidity range  25˚C to 70˚C, 90% rh or less (noncondensing)  
Dimensions (maximum)  430 (440)(16.93''(17.32'')) W × 129.2 (155)(5.09''(6.10'')) H × 550 (620)(21.65''(24.41'')) D mm  
Weight  Approx. 26 kg (57.32 Ib)  Approx. 27 kg (59.52 Ib)  Approx. 25 kg (55.12 Ib)  Approx. 24 kg (52.91 Ib) 
Item  PAT16050T  PAT25032T  PAT35022.8T  PAT50016T  PAT65012.3T  PAT8509.4T  
Input  Nominal input rated voltage  Threephase 200 to 240 VAC, 5060 Hz  
Input voltage range/Input frequency range  180 V to 250 V / 47Hz to 63 Hz  
Efficiency  85% (min) [at input voltage of 200 VAC and rated load]  
Power factor  0.95 (typical) [at input voltage of 200 VAC and rated load]  
Input current  32 A (max) [rated load]  
Inrush current  100 A peak (max)  
Input power  10kVA (max)  
Output  Rating  Rated output power  8 kW  
Rated output voltage  160.0 V  250.0 V  350.0 V  500.0 V  650.0 V  850.0 V  
Rated output current  50.0 A  32.0 A  22.8 A  16.0 A  12.3 A  9.4 A  
Constant voltage  Setting accuracy  ± (0.2% of rating +50 mV)  
Max setting voltage  105% of rating  
Line requlation  ± (0.05% of rating +5 mV)  
Load requlation  ± (0.1% of rating +5 mV)  
Transient response time  5 ms (with sensing at external output, at an instantaneous change in the load current from 50% to 100%)  
Ripple noise  350 mVpp  450 mVpp  450 mVpp  600 mVpp  600 mVpp  600 mVpp  
When the measurement frequency band is 10 Hz to 20 MHz  
30 mVrms  50 mVrms  50 mVrms  100 mVrms  100 mVrms  100 mVrms  
When the measurement frequency band is 5 Hz to 1 MHz  
Raise time  100 ms (rated load)/100 ms (no load)  
Fall time  100 ms (rated load)/2000 ms (no load)  200 ms (rated load)/ 4000 ms (no load)  
Temperature coefficient  100 ppm/˚C (max) [with external analog control]  
Constant current*  Setting accuracy  ± (0.5% of rating +50 mA)  ± (1% of rating +100 mA)  
Max setting current  105% of rating  
Line requlation  ± (0.1% of rating +30 mA)  
Load requlation  ± (0.2% of rating +30 mA)  
Ripple noise  200 mArms  200 mArms  200 mArms  200 mArms  150 mArms  120 mArms  
When the measurement frequency band is 5 Hz to 1 MHz  
Temperature coefficient  200 ppm/˚C (typ) [with external analog control]  
OUTPUT ON/OFF delay  OFF. 0.1 to 10.0 s (resolution: 0.1 s)  
Voltage display  Maximum display  999.9  
Error  ± (0.2% of reading +5 digits) at 23˚C ±5˚C  
Current display  Maximum display  999.9  99.99  
Error  ± (0.5% of reading +5 digits) at 23˚C ±5˚C  
Protection function  Overvoltage protection (OVP) / Overcurrent protection (OCP) / Overheat protection (OHP) /Input open phase protection (PHASE) / Fan error protection (FAN) / Misconnection protection (SENSE) / Breeder circuit overheat protection (BOHP) / Shutdown (SD)  
External analog control  OUTPUT ON/OFF control, etc.  OUTPUT ON/OFF, SHUTDOWN  
Constant voltage, external voltage control  0% to 100% of the rated output voltage at 0 to 10 V  
Constant voltage, external resistance control  0% to 100% or 100% to 0% of the rated output voltage at 0 Wto 10 kW  
Constant current, external voltage control  0% to 100% of tared output current at 0 to 10 V  
Constant current, external resistance control  0% to 100% or 100% to 0% of rated output currenn at 0 Wto 10 kW  
Monitor output  Output voltage  10.00 V ±0.25 V at rated voltage output  
0.00 V ±0.25 V at 0 V output  
Output current  10.00 V ±0.25 V at rated current output  
0.00 V ±0.25 V at 0 A current  
Status output  OUT ON, CV, CC, ALARM, POWER ON, POWER OFF, insulated open collector  
Remote control  Equipped with RS232C interface as standard. SCPI commands, up to 38,400 bps  
Operating temperature/humidity range  0˚C to 50˚C, 20% to 85% rh  
Storage temperature/humidity range  25˚C to 70˚C, 90% rh or less (noncondensing)  
Dimensions (maximum)  430 (440)(16.93''(17.32'')) W × 129.2 (155)(5.09''(6.10'')) H × 550 (620)(21.65''(24.41'')) D mm  
Weight  Approx. 24 kg (52.91 Ib)  Approx. 23 kg (50.71 Ib)  Approx. 22 kg (48.50 Ib)  Approx. 23 kg (50.71 Ib) 
4 kW Type Specifications
Item  PAT20200T  PAT40100T  PAT6067T  PAT16025T  
Input  Nominal input rated voltage  Singlephase/threephase 200 to 240 VAC, 5060 Hz  
Input voltage range/Input frequency range  180 V to 250 V / 47 Hz to 63 Hz  
Efficiency  84% (min)  85% (min) [at input voltage of 200 VAC and rated load]  
Power factor  0.95 (typical) [at input voltage of 200 VAC andratedload] s  
Input current  Singlephase 22 A (max) [at 3 kW load]/threephase 17 A (max) [at rated load]  
Inrush current  50 A peak (max)  
Input power  Singlephase 4 kVA (max) [at 3 kW load]/threephase 5 kVA (max) [at rated load]  
Output  Rating  Rated output power  4 kW(threephase input mode) / 3 kW(singlephase input mode)  
Rated output voltage  20.00 V  40.00 V  60.00 V  160.0 V  
Rated output current  200.0 A  100.0 A  67.00 A  25.00 A  
Constant voltage  Setting accuracy  ± (0.2% of rating +50 mV)  
Max setting voltage  105% of rating  
Line requlation  ± (0.05% of rating +5 mV)  
Load requlation  ± (0.1% of rating +5 mV)  
Transient response time  5 ms (at instantaneous change in load current from 50% to 100%)  
Ripple noise  100 mVpp  300m Vpp  350 mVpp  350 mVpp  
When the measurement frequency band is 10 Hz to 20 MHz  
10 mVrms  30 mVrms  30 mVrms  30 mVrms  
When the measurement frequency band is 5 Hz to 1 MHz  
Raise time  100 ms (rated load)/100 ms (no load)  
Fall time  100 ms (rated load)/2000 ms (no load)  
Temperature coefficient  100 ppm/˚C (max) [with external analog control]  
Constant current*  Setting accuracy  ± (0.5% of rating +50 mA)  
Max setting current  105% of rating × 75% (with singlephase input) / 105% of rating (with threephase input)  
Line requlation  ± (0.1% of rating +30 mA)  
Load requlation  ± (0.2% of rating +30 mA)  
Ripple noise  400 mArms  300 mArms  250 mArms  200 mArms  
When the measurement frequency band is 5 Hz to 1 MHz  
Temperature coefficient  200 ppm/˚C (typ) [with external analog control]  
OUTPUT ON/OFF delay  OFF. 0.1 to 10.0 s (resolution: 0.1 s)  
Voltage display  Maximum display  99.99  999.9  
Error  ± (0.2% of reading +5 digits) at 23˚C ±5˚C  
Current display  Maximum display  999.9  99.99  
Error  ± (0.5% of reading +5 digits) at 23˚C ±5˚C  
Protection function  Overvoltage protection (OVP) / Overcurrent protection (OCP) / Overheat protection (OHP) / Input open phase protection (PHASE) / Fan error protection (FAN) /Misconnection protection (SENSE) / Breeder circuit overheat protection (BOHP) / Shutdown (SD)  
External analog control  OUTPUT ON/OFF control, etc.  OUTPUT ON/OFF, SHUTDOWN  
Constant voltage, external voltage control  0% to 100% of the rated output voltage at 0 to 10 V  
Constant voltage, external resistance control  0% to 100% or 100% to 0% of the rated output voltage at 0 Wto 10 kW  
Constant current, external voltage control  0% to 100% of tared output current at 0 to 10 V  
Constant current, external resistance control  0% to 100% or 100% to 0% of rated output currenn at 0 Wto 10 kW  
Monitor output  Output voltage  10.00 V ±0.25 V at rated voltage output  
0.00 V ±0.25 V at 0 V output  
Output current  10.00 V ±0.25 V at rated current output  
0.00 V ±0.25 V at 0 A current  
Status output  OUT ON, CV, CC, ALARM, POWER ON, POWER OFF, insulated open collector  
Remote control  Equipped with RS232C interface as standard. SCPI commands, up to 38,400 bps  
Operating temperature/humidity range  0˚C to 50˚C, 20% to 85% rh  
Storage temperature/humidity range  25˚C to 70˚C, 90% rh or less (noncondensing)  
Dimensions (maximum)  430 (440)(16.93''(17.32'')) W × 129.2 (155)(5.09''(6.10'')) H × 550 (620)(21.65''(24.41'')) D mm  
Weight  Approx. 20 kg(44.09 Ib)  Approx. 19 kg(41.89 Ib)  Approx. 18 kg(39.68 Ib) 
OPTIONS
PC01PAT
Parallel operation cable for PAT series length:250 mm
AC84P4MM6C
Input Power cable, Threephase, 4conductor 4m M6
KRB3TOS
Option(s) for Electrical Safety Testers. Bracket (Inch size) for TOS9200, TOS9201, TOS8870A, PAM(2kW type), PCR1000M, LZ1205W, PLZ4W/4WH, PWR, PAT, PBZ
KRB150TOS
Option(s) for Electrical Safety Testers. Bracket (Metric size) for TOS9200, TOS9201, TOS8870A, PAM(2kW type), PCR1000M, PLZ1205W, PLZ4W/4WH, PWR, PAT, PBZ
OP01PAT
Power switch guard
VS01
Vertical stand
Related Products

Kikusui PWR Series
Variableswitching Multi Range DC Power Supply (Switching system, CV/CC) : 9 Models

Kikusui Pav Serisi DC Güç Kaynakları
Smart Variableswitching DC Power Supply (Switching system, CV/CC) : 32 Models

Kikusui PWX Series
1U Multi Range Programmable DC Power Supply (Switching system, CV/CC) : 9 Models
Ürünler
 ►Solar Simülatör
 ►Sayaç Test Cihazları
 ►Motor Test Cihazları
 Yüksek Gerilim Bolücüler
 Yüksek Akım Kaynağı
 Megohm meter
 ►Toprak Megerleri
 ►İzolasyon/Yüksek Gerilim (DelinmeHipot)/Kaçak Akım/Topraklama Test Cihazları
 ►Güç Analizörleri
 ►Güneş Ölçümü Test Cihazları
 ►Trafo Test Cihazları
 ►MiliMikro Ohm Metreler
 ►Şalt Sahası Test Cihazları
 ►Kablo Test Cihazları
 ►Termal Kameralar
 ►Kalibratörler
 ►LCR Metreler
 ►Osiloskoplar
 ►Data Loggerlar
 ▼Güç Kaynakları
 ►PD (Kısmi Boşalma) Tespit Cihazları
 ►Akü Test Cihazları
 ►Diğer Ürünler
 ►Deney Setleri / Eğitim Setleri
 Yeşil Enerji Eğitim Setleri
 Akıllı Bina Deney Setleri (EIB/KNX)
 Fonksiyon/Arbitary Dalga Formu Jernaratörü
 Elektrik Tesisatı Eğitim Setleri
 Flicker Eğitim Seti
 Temel Elektrik Mühendisliği/Elektronik/Analog teknolojisi
 Güç Elektroniği / Kontrol Mühendisliği Deney Setleri
 Elektrik Makineleri Eğitim Seti
 Haberleşme / Radyo Mühendisliği Eğitim Setleri
 Dijital ve Otomasyon Teknolojisi Eğitim Setleri
 Kontaktör Kontrol Eğitim Setleri
 Mekatronik Kompakt Sistem Eğitim Seti
 ►İş Güvenliği Ekipmanları ve Test Cihazları
 ►Laboratuvar & Çevresel Ölçüm Cihazları
 ►Multimetreler
 ►Pens Ampermetreler
 ►Röle Test Cihazları
 ►Voltaj Dedektörleri
 ►Yarı İletken Test ve Üretim Sistemleri
 ►Piranometreler
 ►LPCVD/PECVD
 ►PAR / FAR Ölçümü